Maintenance Requirements for Anesthesia Machine
1. After the end of the daily operation, the flow sensor module and the patient's breathing circuit must be removed for maintenance. It is recommended to use a high temperature and high-pressure method to disinfect the silicone suction line.
2. Wipe the dirt on each surface of the main unit with a neutral detergent.
3. Use as many flow sensors as possible, alternate between different patients or long-term surgery to extend the life of the sensor. The flow sensor needs to be dried using a low oxygen flow of pure oxygen to keep it dry. If the water is still not working properly after removing the water, and the flow sensor is replaced, the original flow sensor is scrapped.
4. Regularly check that the detection and alarm functions of Portable Anesthesia Vaporizer work effectively.
5. When disinfecting indoors, be sure to keep the anesthesia machine away from direct ultraviolet rays and prevent the aging of accessories such as accelerating pipes.
6. The battery built in the main ventilator is fully charged and discharged once a month and recharged to maintain the good performance of the battery.
- From the Dolphinmed office center news, through the inspection and acceptance by by State Administration of Work, Dolphinmed Work Safety Standardization got the got recognized by State Administration of Work Safety. Following is the certificate.