Elimination Of Simple Malfunction Of Anesthesia Machine
As an Anesthesia Workstation Manufacturer, we have the following methods for the elimination of simple malfunctions of anesthesia machines.
1. The anesthesia machine has a leaking sound, which may be due to a broken pipe or loose joints, and there are warnings such as Pressure Low, Peak Low, or no warning. In general, when you hear a leak, mainly a leak in a high-pressure part (such as a central gas supply fitting or gas supply to the main switch of the machine), you can check the piping or fittings in the relevant part. Because the leaking part is mainly medical gas (air) and oxygen, if it can not be eliminated in time, it should be repaired or replaced after the patient's operation is completed according to the site conditions.
2. When the air bag is lowered during the test, it may be because the breathing circuit is leaking. It is recommended to check the pressure relief valve, re-adjust the position, or replace the pressure relief valve and the breathing air bag set.
3. The respirator cannot correctly determine the position of the Bag/Vent switch. The possible cause is that the respirator or carbon dioxide absorber is faulty. It should be changed to manual breathing and contact the maintenance personnel for maintenance.
4. There is a large amount of air leakage in the breathing circuit (manual mode). The possible reason is that the manual ball is not connected properly, or the sodium and lime absorption tank is not properly installed. It is recommended to confirm that the manual ball connecting pipe is properly connected or reinstall the sodium lime absorption tank.
5. The power supply often loses power when the power is off. The possible reason is that the load connected to the power output device is too large. It is recommended to check whether the power output device is short-circuited or used in a distributed outlet.
Our company provides Portable Anesthesia Machine. Welcome to contact us.
- From the Dolphinmed office center news, through the inspection and acceptance by by State Administration of Work, Dolphinmed Work Safety Standardization got the got recognized by State Administration of Work Safety. Following is the certificate.